state), and can achieve sub-nanometer resolution. In its simplest form, the technique is applicable to thin films with thickness of less than a nanometer to several micrometers. Most models assume the sample is composed of a small number of discrete, well-defined layers that are optically homogeneous and isotropic. Violation of these assumptions requires more advanced variants of the technique (see below). Methods of immersion or multiangular ellipsometry are applied to find the optical constants of the material with rough sample surface or presence of inhomogeneous media. New methodological approaches allow the use of reflection ellipsometry to measure physical and technical characteristics of gradient elements in case the surface layer of the optical detail is inhomogeneous. Experimental setup Electromagnetic radiation is emitted by a light source and linearly polarized by a polarizer. It can pass through an optional compensator (retarder, quarter wave plate) and falls onto the sample. After reflection the radiation passes a compensator (optional) and a second polarizer, which is called an analyzer, and falls into the detector. Instead of the compensators, some ellipsometers use a phase-modulator in the path of the incident light beam. Ellipsometry is a specular optical technique (the angle of incidence equals the angle of reflection). The incident and the reflected beam span the plane of incidence. Light which is polarized parallel to this plane is named p-polarized. A polarization direction perpendicular is called s-polarized (s-polarised), accordingly. The "s" is contributed from the German "" (perpendicular). Data acquisition Ellipsometry measures the complex reflectance ratio of a system, which may be parametrized by the amplitude component and the phase difference . The polarization state of the light incident upon the sample may be decomposed into an s and a p component (the s component is oscillating perpendicular to the plane of incidence and parallel to the sample surface, and the p component is oscillating parallel to the plane of incidence). The amplitudes of the s and p components, after reflection and normalized to their initial value, are denoted by and respectively. The angle of incidence is chosen close to the Brewster angle of the sample to ensure a maximal difference in and . Ellipsometry measures the complex reflectance ratio (a complex quantity), which is the ratio of over : Thus, is the amplitude ratio upon reflection, and is the phase shift (difference). (Note that the right side of the equation is simply another way to represent