Accelerator mass spectrometry (AMS) is a form of mass spectrometry that accelerates ions to extraordinarily high kinetic energies before mass analysis. The special strength of AMS among the mass spectrometric methods is its power to separate a rare isotope from an abundant neighboring mass ("abundance sensitivity", e.g. 14C from 12C). The method suppresses molecular isobars completely and in many cases can separate atomic isobars (e.g. 14N from 14C) also. This makes possible the detection of naturally occurring, long-lived radio-isotopes such as 10Be, 36Cl, 26Al and 14C. Their typical isotopic abundance ranges from 10−12 to 10−18. AMS can outperform the competing technique of decay counting for all isotopes where the half-life is long enough. Other advantages of AMS include its short measuring time as well as its ability to detect atoms in extremely small samples. The method Generally, negative ions are created (atoms are ionized) in an ion source. In fortunate cases, this already allows the suppression of an unwanted isobar, which does not form negative ions (as 14N in the case of 14C measurements). The pre-accelerated ions are usually separated by a first mass spectrometer of sector-field type and enter an electrostatic "tandem accelerator". This is a large nuclear particle accelerator based on the principle of a Tandem van de Graaff Accelerator operating at 0.2 to many million volts with two stages operating in tandem to accelerate the particles. At the connecting point between the two stages, the ions change charge from negative to positive by passing through a thin layer of matter ("stripping", either gas or a thin carbon foil). Molecules will break apart in this stripping stage. The complete suppression of molecular isobars (e.g. 13CH− in the case of 14C measurements) is one reason for the exceptional abundance sensitivity of AMS. Additionally, the impact strips off several of the ion's electrons, converting it into a positively charged ion. In the second half of the accelerator, the now positively charged ion is accelerated away from the highly positive centre of the electrostatic accelerator which previously attracted the negative ion. When the ions leave the accelerator they are positively charged and are moving at several percent of the speed of light. In the second stage of mass spectrometer, the fragments from the molecules are separated from the ions of interest. This spectrometer may consist of magnetic or electric sectors, and so-called velocity selectors, which utilizes both electric fields and magnetic