A resuscitator is a device using positive pressure to inflate the lungs of an unconscious person who is not breathing, in order to keep them oxygenated and alive. There are three basic types: a manual version (also known as a bag valve mask) consisting of a mask and a large hand-squeezed plastic bulb using ambient air, or with supplemental oxygen from a high-pressure tank. The second type is the Expired Air or breath powered resuscitator. The first appearance of the second type was the Brooke Airway introduced in 1957. The third type is an oxygen powered resuscitator. These are driven by pressurized gas delivered by a regulator, and can either be automatic or manually controlled. The most popular type of gas powered resuscitator are Time Cycled, Volume Constant Ventilators. In the early days of pre-hospital emergency services, pressure cycled devices like the Pulmotor were popular but yielded less than satisfactory results. One of the first modern resuscitation ventilators was the HARV, later called the PneuPac 2R or Yellow Box. Most modern resuscitators are designed to allow the patient to breathe on his own should he recover the ability to do so. All resuscitation devices should be able to deliver >85% oxygen when a gas source is available. History Resuscitators began in 1907 when Heinrich Dräger, owner of the Drägerwerk AG Company, produced the "Pulmotor" Resuscitator. Considered to be the first practical device for delivering oxygen to unconscious patients or patients in respiratory distress, the Pulmotor influenced resuscitators for many years. When ambulance services began to form in major cities around the world, such as in London, New York and Los Angeles, Emergency medical services or EMS was developed. In these early days, perhaps the most advanced piece of equipment carried on these ambulances were devices for delivering supplemental oxygen to patients in respiratory distress. The Pulmotor and later models, such as the Emerson Resuscitator, utilized heavy tanks of oxygen to power a device which forced air into the patient's lungs. While better than no oxygen at all, these old units were problematic. Aside from often failing to sense obstructions in the airway, the Emerson, and to a lesser degree the Pulmotor, were large, bulky and heavy. The Emerson Resuscitator required two strong men to carry it from the ambulance to the victim. Perhaps the greatest defect, however, was the fact that these units "cycled". Cycling was a feature that was