An LCR meter is a type of electronic test equipment used to measure the inductance (L), capacitance (C), and resistance (R) of an electronic component. In the simpler versions of this instrument the impedance was measured internally and converted for display to the corresponding capacitance or inductance value. Readings should be reasonably accurate if the capacitor or inductor device under test does not have a significant resistive component of impedance. More advanced designs measure true inductance or capacitance, as well as the equivalent series resistance of capacitors and the Q factor of inductive components. Operation Usually the device under test (DUT) is subjected to an AC voltage source. The meter measures the voltage across and the current through the DUT. From the ratio of these the meter can determine the magnitude of the impedance. The phase angle between the voltage and current is also measured in more advanced instruments; in combination with the impedance, the equivalent capacitance or inductance, and resistance, of the DUT can be calculated and displayed. The meter must assume either a parallel or a series model for these two elements. An ideal capacitor has no characteristics other than capacitance, but there are no physical ideal capacitors. All real capacitors have a little inductance, a little resistance, and some defects causing inefficiency. These can be seen as inductance or resistance in series with the ideal capacitor or in parallel with it. And so likewise with inductors. Even resistors can have inductance (especially if they are wire wound types) and capacitance as a consequence of the way they are constructed. The most useful assumption, and the one usually adopted, is that LR measurements have the elements in series (as is necessarily the case in an inductor's coil) and that CR measurements have the elements in parallel (as is necessarily the case between a capacitor's 'plates'). Leakage is a special case in capacitors, as the leakage is necessarily across the capacitor plates, that is, in series. An LCR meter can also be used to measure the inductance variation with respect to the rotor position in permanent magnet machines. (However, care must be taken, as some LCR meters will be damaged by the generated EMF produced by turning the rotor of a permanent-magnet motor; in particular those intended for electronic component measurements.) Handheld LCR meters typically have selectable test frequencies of 100 Hz, 120 Hz, 1 kHz, 10 kHz, and