Analog multimeters had the precision and reading accuracy limitations described above, and so were not built to provide the same accuracy as digital instruments. Analog meters were intuitive where the trend of a measurement was more important than an exact value obtained at a particular moment. A change in angle or in a proportion was easier to interpret than a change in the value of a digital readout. For this reason, some digital multimeters additionally have a bar graph as a second display, typically with a more rapid sampling rate than used for the primary readout. These fast sampling rate bar graphs have a superior response than the physical pointer of analog meters, obsoleting the older technology. With rapidly fluctuating DC, AC or a combination of both, advanced digital meters were able to track and display fluctuations better than analog meters whilst also having the ability to separate and simultaneously display DC and AC components. Analog meter movements are inherently more fragile physically and electrically than digital meters. Many analog multimeters feature a range switch position marked "off" to protect the meter movement during transportation which places a low resistance across the meter movement, resulting in dynamic braking. Meter movements as separate components may be protected in the same manner by connecting a shorting or jumper wire between the terminals when not in use. Meters which feature a shunt across the winding such as an ammeter may not require further resistance to arrest uncontrolled movements of the meter needle because of the low resistance of the shunt. The meter movement in a moving pointer analog multimeter is practically always a moving-coil galvanometer of the d'Arsonval type, using either jeweled pivots or taut bands to support the moving coil. In a basic analog multimeter the current to deflect the coil and pointer is drawn from the circuit being measured; it is usually an advantage to minimize the current drawn from the circuit, which implies delicate mechanisms. The sensitivity of an analog multimeter is given in units of ohms per volt. For example, a very low-cost multimeter with a sensitivity of 1,000 Ω/V would draw 1 mA from a circuit at full-scale deflection. More expensive, (and mechanically more delicate) multimeters typically have sensitivities of 20,000 ohms per volt and sometimes higher, with 50,000 ohms per volt (drawing 20 microamperes at full scale) being about the upper limit for a portable, general purpose,