the thickness achievable is between 10 and 100 μm. The device operates using a cutting action of an infrared laser. As the laser emits a radiation in the near infrared, in this wavelength regime the laser can interact with biological materials. Through sharp focusing of the probe within the sample, a focal point of very high intensity, up to TW/cm2, can be achieved. Through the non-linear interaction of the optical penetration in the focal region a material separation in a process known as photo-disruption is introduced. By limiting the laser pulse durations to the femtoseconds range, the energy expended at the target region is precisely controlled, thereby limiting the interaction zone of the cut to under a micrometre. External to this zone the ultra-short beam application time introduces minimal to no thermal damage to the remainder of the sample. The laser radiation is directed onto a fast scanning mirror-based optical system, which allows three-dimensional positioning of the beam crossover, whilst allowing beam traversal to the desired region of interest. The combination of high power with a high raster rate allows the scanner to cut large areas of sample in a short time. In the laser microtome the laser-microdissection of internal areas in tissues, cellular structures, and other types of small features is also possible. Knives The selection of microtome knife blade profile depends upon the material and preparation of the samples, as well as the final sample requirements (e.g. cut thickness and quality). Design and cut types Generally, knives are characterized by the profile of the knife blade, which falls under the categories of planar concave, wedge shaped or chisel shaped designs. Planar concave microtome knives are extremely sharp, but are also very delicate and are therefore only used with very soft samples. The wedge profile knives are somewhat more stable and find use in moderately hard materials, such as in epoxy or cryogenic sample cutting. Finally, the chisel profile with its blunt edge, raises the stability of the knife, whilst requiring significantly more force to achieve the cut. For ultramicrotomes, glass and diamond knives are required, the cut breadth of the blade is therefore on the order of a few millimetres and is therefore significantly smaller than for classical microtome knives. Glass knives are usually manufactured by the fracture of glass bars using special "knife-maker" fracturing devices. Glass knives may be used for initial sample preparations even where diamond