laboratory conditions and with highly trained experts, it is possible to achieve small (a few tenths to a few percent) errors in measurements. However, in many practical situations, there is the likelihood of errors on the order of 10 percent Several types of error are at play when taking physical measurements. The three basic types of error noted as the limiting factors of accuracy of measurement are random, systematic, and periodic errors Random errors are variations about that mean. In the case of spectroradiometric measurements, this could be thought of as noise from the detector, internal electronics, or the light source itself. Errors of this type can be combated by longer integration times or multiple scans. Systematic errors are offsets to the predicted "correct" value. Systematic errors generally occur due to the human component of these measurements, the device itself, or the setup of the experiment. Things such as calibration errors, stray light, and incorrect settings, are all potential issues. Periodic errors arise from recurrent periodic or pseudo-periodic events. Variations in temperature, humidity, air-motion, or AC interference could all be categorized as periodic error. In addition to these generic sources of error, a few of the more specific reasons for error in spectroradiometry include: The multidimensionality of the measurement. The output signal is dependent on several factors, including magnitude of measured flux, its direction, its polarization, and its wavelength distribution. The inaccuracy of measuring instruments, as well as the standards used to calibrate said instruments, cascaded to create a larger error throughout the entire measurement process, and The proprietary techniques for reducing multidimensionality and device instability error. Gamma-scientific, a California-based manufacturer of light measurement devices, lists seven factors affecting the accuracy and performance of their spectroradiometers, due to either the system calibration, the software and power supply, the optics, or the measurement engine itself. Definitions Stray light Stray light is unwanted wavelength radiation reaching the incorrect detector element. It generates erroneous electronic counts not related to designed spectral signal for the pixel or element of the detector array. It can come from light scatter and reflection of imperfect optical elements as well as higher order diffraction effects. The second order effect can be removed or at least dramatically reduced, by installing order sorting filters before the detector. A Si detectors sensitivity to visible and NIR is nearly an order of magnitude larger than that in the UV range. This means