primary flux is no longer canceled by secondary flux, smaller current transformers may not actually incur dangerous voltages when operating nominally. Faster current transients from loads being switched on etc. can however still induce dangerous voltage levels due to high current slope. Accuracy The accuracy of a CT is affected by a number of factors including: Burden Burden class/saturation class Rating factor Load External electromagnetic fields Temperature Physical configuration The selected tap, for multi-ratio CTs Phase change Capacitive coupling between primary and secondary Resistance of primary and secondary Core magnetizing current Accuracy classes for various types of measurement and at standard loads in the secondary circuit (burdens) are defined in IEC 61869-1 as classes 0.1, 0.2s, 0.2, 0.5, 0.5s, 1 and 3. The class designation is an approximate measure of the CT's accuracy. The ratio (primary to secondary current) error of a Class 1 CT is 1% at rated current; the ratio error of a Class 0.5 CT is 0.5% or less. Errors in phase are also important, especially in power measuring circuits. Each class has an allowable maximum phase error for a specified load impedance. Current transformers used for protective relaying also have accuracy requirements at overload currents in excess of the normal rating to ensure accurate performance of relays during system faults. A CT with a rating of 2.5L400 specifies with an output from its secondary winding of twenty times its rated secondary current (usually ) and 400 V (IZ drop) its output accuracy will be within 2.5 percent. Burden The secondary load of a current transformer is termed the "burden" to distinguish it from the primary load. The burden in a CT metering electrical network is largely resistive impedance presented to its secondary winding. Typical burden ratings for IEC CTs are 1.5 VA, 3 VA, 5 VA, 10 VA, 15 VA, 20 VA, 30 VA, 45 VA and 60 VA. ANSI/IEEE burden ratings are B-0.1, B-0.2, B-0.5, B-1.0, B-2.0 and B-4.0. This means a CT with a burden rating of B-0.2 will maintain its stated accuracy with up to 0.2 Ω on the secondary circuit. These specification diagrams show accuracy parallelograms on a grid incorporating magnitude and phase angle error scales at the CT's rated burden. Items that contribute to the burden of a current measurement circuit are switch-blocks, meters and intermediate conductors. The most common cause of excess burden impedance is the conductor between the meter