proves that the redox species that gives the peak is actually immobilised). The effect of increasing the scan rate can be used to measure the rate of interfacial electron transfer and/or the rates of reactions that are coupltransfer. This technique has been useful to study redox proteins, some of which readily adsorb on various electrode materials, but the theory for biological and non-biological redox molecules is the same (see the page about protein film voltammetry). Experimental setup CV experiments are conducted on a solution in a cell fitted with electrodes. The solution consists of the solvent, in which is dissolved electrolyte and the species to be studied. The cell A standard CV experiment employs a cell fitted with three electrodes: reference electrode, working electrode, and counter electrode. This combination is sometimes referred to as a three-electrode setup. Electrolyte is usually added to the sample solution to ensure sufficient conductivity. The solvent, electrolyte, and material composition of the working electrode will determine the potential range that can be accessed during the experiment. The electrodes are immobile and sit in unstirred solutions during cyclic voltammetry. This "still" solution method gives rise to cyclic voltammetry's characteristic diffusion-controlled peaks. This method also allows a portion of the analyte to remain after reduction or oxidation so that it may display further redox activity. Stirring the solution between cyclic voltammetry traces is important in order to supply the electrode surface with fresh analyte for each new experiment. The solubility of an analyte can change drastically with its overall charge; as such it is common for reduced or oxidized analyte species to precipitate out onto the electrode. This layering of analyte can insulate the electrode surface, display its own redox activity in subsequent scans, or otherwise alter the electrode surface in a way that affects the CV measurements. For this reason it is often necessary to clean the electrodes between scans. Common materials for the working electrode include glassy carbon, platinum, and gold. These electrodes are generally encased in a rod of inert insulator with a disk exposed at one end. A regular working electrode has a radius within an order of magnitude of 1 mm. Having a controlled surface area with a well-defined shape is necessary for being able to interpret cyclic voltammetry results. To run cyclic voltammetry experiments at very high scan rates a regular working electrode is insufficient. High scan rates create peaks with