large currents and increased resistances, which result in distortions. Ultramicroelectrodes can be used to minimize the current and resistance. The counter electrode, also known as the auxiliary or second electrode, can be any material that conducts current easily, will not react with the bulk solution, and has a surface area much larger than the working electrode. Common choices are platinum and graphite. Reactions occurring at the counter electrode surface are unimportant as long as it continues to conduct current well. To maintain the observed current the counter electrode will often oxidize or reduce the solvent or bulk electrolyte. Solvents CV can be conducted using a variety of solutions. Solvent choice for cyclic voltammetry takes into account several requirements. The solvent must dissolve the analyte and high concentrations of the supporting electrolyte. It must also be stable in the potential window of the experiment with respect to the working electrode. It must not react with either the analyte or the supporting electrolyte. It must be pure to prevent interference. Electrolyte The electrolyte ensures good electrical conductivity and minimizes iR drop such that the recorded potentials correspond to actual potentials. For aqueous solutions, many electrolytes are available, but typical ones are alkali metal salts of perchlorate and nitrate. In nonaqueous solvents, the range of electrolytes is more limited, and a popular choice is tetrabutylammonium hexafluorophosphate. Related potentiometric techniques Potentiodynamic techniques also exist that add low-amplitude AC perturbations to a potential ramp and measure variable response in a single frequency (AC voltammetry) or in many frequencies simultaneously (potentiodynamic electrochemical impedance spectroscopy). The response in alternating current is two-dimensional, characterized by both amplitude and phase. These data can be analyzed to determine information about different chemical processes (charge transfer, diffusion, double layer charging, etc.). Frequency response analysis enables simultaneous monitoring of the various processes that contribute to the potentiodynamic AC response of an electrochemical system. Whereas cyclic voltammetry is not hydrodynamic voltammetry, useful electrochemical methods are. In such cases, flow is achieved at the electrode surface by stirring the solution, pumping the solution, or rotating the electrode as is the case with rotating disk electrodes and rotating ring-disk electrodes. Such techniques target steady state conditions and produce waveforms that appear the same when scanned in either the positive or negative directions, thus limiting them to linear sweep voltammetry. Applications Cyclic voltammetry (CV) has become an important and widely used electroanalytical technique in many areas