of the current in a spontaneous gas discharge in intersecting electric and magnetic fields. UHV pressures are measured with an ion gauge, either of the hot filament or inverted magnetron type. Leak rate In any vacuum system, some gas will continue to escape into the chamber over time and slowly increase the pressure if it is not pumped out. This leak rate is usually measured in mbar L/s or torr L/s. While some gas release is inevitable, if the leak rate is too high, it can slow down or even prevent the system from reaching low pressure. There are a variety of possible reasons for an increase in pressure. These include simple air leaks, virtual leaks, and desorption (either from surfaces or volume). A variety of methods for leak detection exist. Large leaks can be found by pressurizing the chamber, and looking for bubbles in soapy water, while tiny leaks can require more sensitive methods, up to using a tracer gas and specialized Helium mass spectrometer. Outgassing Outgassing is a problem for UHV systems. Outgassing can occur from two sources: surfaces and bulk materials. Outgassing from bulk materials is minimized by selection of materials with low vapor pressures (such as glass, stainless steel, and ceramics) for everything inside the system. Materials which are not generally considered absorbent can outgas, including most plastics and some metals. For example, vessels lined with a highly gas-permeable material such as palladium (which is a high-capacity hydrogen sponge) create special outgassing problems. Outgassing from surfaces is a subtler problem. At extremely low pressures, more gas molecules are adsorbed on the walls than are floating in the chamber, so the total surface area inside a chamber is more important than its volume for reaching UHV. Water is a significant source of outgassing because a thin layer of water vapor rapidly adsorbs to everything whenever the chamber is opened to air. Water evaporates from surfaces too slowly to be fully removed at room temperature, but just fast enough to present a continuous level of background contamination. Removal of water and similar gases generally requires baking the UHV system at while vacuum pumps are running. During chamber use, the walls of the chamber may be chilled using liquid nitrogen to reduce outgassing further. Bake-out In order to reach low pressures, it is often useful to heat the entire system above for many hours (a process known as bake-out)