diffusion layer at the electrode surface. Characterization The utility of cyclic voltammetry is highly dependent on the analyte being studied. The analyte has to be redox active within the potential window to be scanned. The analyte is in solution Reversible couples Often the analyte displays a reversible CV wave (such as that depicted in Figure 1), which is observed when all of the initial analyte can be recovered after a forward and reverse scan cycle. Although such reversible couples are simpler to analyze, they contain less information than more complex waveforms. The waveform of even reversible couples is complex owing to the combined effects of polarization and diffusion. The difference between the two peak potentials (Ep), ΔEp, is of particular interest. ΔEp = Epa - Epc > 0 This difference mainly results from the effects of analyte diffusion rates. In the ideal case of a reversible 1e- couple, ΔEp is 57 mV and the full-width half-max of the forward scan peak is 59 mV. Typical values observed experimentally are greater, often approaching 70 or 80 mV. The waveform is also affected by the rate of electron transfer, usually discussed as the activation barrier for electron transfer. A theoretical description of polarization overpotential is in part described by the Butler–Volmer equation and Cottrell equation equations. In an ideal system the relationships reduces to for an n electron process. Focusing on current, reversible couples are characterized by ipa/ipc = 1. When a reversible peak is observed, thermodynamic information in the form of a half cell potential E01/2 can be determined. When waves are semi-reversible (ipa/ipc is close but not equal to 1), it may be possible to determine even more specific information (see electrochemical reaction mechanism). Nonreversible couples Many redox processes observed by CV are quasi-reversible or non-reversible. In such cases the thermodynamic potential E01/2 is often deduced by simulation. The irreversibility is indicated by ipa/ipc ≠ 1. Deviations from unity are attributable to a subsequent chemical reaction that is triggered by the electron transfer. Such EC processes can be complex, involving isomerization, dissociation, association, etc. The analyte is adsorbed onto the electrode surface Adsorbed species give simple voltammetric responses: ideally, at slow scan rates, there is no peak separation, the peak width is 90mV for a one-electron redox couple, and the peak current and peak area are proportional to scan rate (observing that the peak current is proportional to scan rate