released from the wearer's the mouth and nose. Surgical masks help reduce exposure of the wearer's saliva and respiratory secretions to others that could otherwise travel up to . Surgical mask also remind wearers not to touch their mouth or nose, which could otherwise transfer viruses and bacteria after having touched a contaminated surface. A surgical mask is not to be confused with a respirator (which is specifically rated for sub-micron particles) and is not certified as such. Surgical masks are not designed to protect the wearer from inhaling airborne bacteria or virus particles and are less effective than respirators, which are designed for this purpose. Collection efficiency of surgical mask filters can range from less than 10% to nearly 90% for different manufacturers’ masks when measured using the test parameters for NIOSH certification. However, a study found that even for surgical masks with "good" filters, 80–100% of subjects failed an OSHA-accepted qualitative fit test, and a quantitative test showed 12–25% leakage. Modern surgical masks are made from paper or other non-woven material and should be discarded after each use. Physical form The design of the surgical masks depends on the intended usages. Usually, the masks are rectangular-shaped with pleats to allow the wearer to expand and curve the mask so it can better cover all the area from the nose to around the chin. The outward-facing side of the mask are typically colored (usually blue, green or yellow) and made thicker, tougher and water-impermeable. The inner layers of the mask are made of three-ply (three layers) melt-blown polymer (most commonly polypropylene) placed between non-woven fabric. The melt-blown material acts as the filter that stops microbes from penetrating and exiting the mask. Some masks have an attached thin polyethylene faceshield (known as "splash shield") to provide additional spray protection over the eye area. A different type of masks, known as "duckbill" masks, uses a trapezoid pouch-like design that has significantly shortened side edges — sometimes none at all — to minimize loose gaps that pathogens can leak past. These masks are typically made to the N95/P2 standards, and are commonly used for clinical situations that demand fine particulates protection, such as tuberculosis. Small strips of foam or thickened fabric are often sewn along the top edge of the mask to help better seal away exhaled water vapors (which can fog up eyewears and faceshields) and soak up excess perspiration dripping