correspond to a specific pressure on the pressure-volume curve and vice versa in any case. Settings on each mechanical ventilator may include respiratory rate, tidal volume, trigger sensitivity, flow rate, waveform, and inspiratory/expiratory ratio. The volume-cycled ventilation includes the volume-control function and delivers a set tidal volume. The pressure is not a fixed number but it varies with resistance and capacitance of the respiratory system. The volume-cycled ventilation is the simplest and most efficient of providing ventilation to a patient's airway compared to other methods of mechanical ventilation. Each inspiratory effort that is beyond the set sensitivity threshold will be accounted for and fixed to the delivery of the corresponding tidal volume. If the patient does not breathe enough, then the volume-cycled ventilation will initiate a breath for the patient to bring up the breathing rate to the minimum respiratory rate. The synchronized intermittent mandatory ventilation (SIMV) is a similar method of mechanical ventilation that also delivers breaths at a fixed rate and volume that corresponds to the patient's breathing. Unlike the Volume-Cycled Ventilation, patient efforts above the fixed rate are unassisted in the synchronized intermittent mandatory ventilation (SIMV). Pressure-cycled ventilation The pressure-cycled ventilation includes pressure control ventilation and pressure support ventilation. Both methods offer a set inspiratory pressure. The tidal volume varies depending on the resistance and elastance of the respiratory system. Pressure-cycled ventilation can help alleviate symptoms in patients with acute respiratory distress syndrome by limiting the distending pressure of the lungs. The pressure control ventilation is specifically a pressure-cycled form of assist-control ventilators. Assist-control ventilators describe a mode of ventilation that maintains a minimum respiratory rate regardless of whether or not the patient initiates a spontaneous breath. Each inspiratory effort that is beyond the sensitivity threshold delivers full pressures support for a fixed inspiratory time. There is maintenance of a minimum respiratory rate. In the pressure support ventilation, the minimum rate is not set. Instead, all breaths are triggered by the patient. The way that the pressure support ventilation works is by assisting the patient with a constant pressure until the patient's inspiratory flow falls below a threshold. The longer, deeper inspiratory flows by the patient will result in a larger tidal volume. This method of mechanical ventilation will help patients assume more work of breathing. Noninvasive positive pressure ventilation (NIPPV) Noninvasive positive pressure ventilation is the delivery of positive pressure ventilation through a tight-fitting mask that