the use of a set of buffer solutions of known activity. Two or more buffer solutions are used in order to accommodate the fact that the "slope" may differ slightly from ideal. To implement this approach to calibration, the electrode is first immersed in a standard solution and the reading on a pH meter is adjusted to be equal to the standard buffer's value. The reading from a second standard buffer solution is then adjusted, using the "slope" control, to be equal to the pH for that solution. Further details, are given in the IUPAC recommendations. When more than two buffer solutions are used the electrode is calibrated by fitting observed pH values to a straight line with respect to standard buffer values. Commercial standard buffer solutions usually come with information on the value at 25 °C and a correction factor to be applied for other temperatures. The pH scale is logarithmic and therefore pH is a dimensionless quantity. P[H] This was the original definition of Sørensen in 1909, which was superseded in favor of pH in 1924. [H] is the concentration of hydrogen ions, denoted [] in modern chemistry, which appears to have units of concentration. More correctly, the thermodynamic activity of in dilute solution should be replaced by []/c0, where the standard state concentration c0 = 1 mol/L. This ratio is a pure number whose logarithm can be defined. However, it is possible to measure the concentration of hydrogen ions directly, if the electrode is calibrated in terms of hydrogen ion concentrations. One way to do this, which has been used extensively, is to titrate a solution of known concentration of a strong acid with a solution of known concentration of strong alkaline in the presence of a relatively high concentration of background electrolyte. Since the concentrations of acid and alkaline are known, it is easy to calculate the concentration of hydrogen ions so that the measured potential can be correlated with concentrations. The calibration is usually carried out using a Gran plot. Thus, the effect of using this procedure is to make activity equal to the numerical value of concentration. The glass electrode (and other ion selective electrodes) should be calibrated in a medium similar to the one being investigated. For instance, if one wishes to measure the pH of a seawater sample, the electrode should be calibrated in a solution resembling seawater in its chemical composition,