unaffected by cup size or arm length. This was apparently confirmed by some early independent experiments, but it was incorrect. Instead, the ratio of the speed of the wind and that of the cups, the anemometer factor, depends on the dimensions of the cups and arms, and can have a value between two and a little over three. Once the error was discovered, all previous experiment involving anemometers had to be repeated. The three-cup anemometer developed by Canadian John Patterson in 1926, and subsequent cup improvements by Brevoort & Joiner of the United States in 1935, led to a cupwheel design with a nearly linear response and an error of less than 3% up to . Patterson found that each cup produced maximum torque when it was at 45° to the wind flow. The three-cup anemometer also had a more constant torque and responded more quickly to gusts than the four-cup anemometer. The three-cup anemometer was further modified by Australian Dr. Derek Weston in 1991 to also measure wind direction. He added a tag to one cup, causing the cupwheel speed to increase and decrease as the tag moved alternately with and against the wind. Wind direction is calculated from these cyclical changes in speed, while wind speed is determined from the average cupwheel speed. Three-cup anemometers are currently the industry standard for wind resource assessment studies and practice. Vane anemometers One of the other forms of mechanical velocity anemometer is the vane anemometer. It may be described as a windmill or a propeller anemometer. Unlike the Robinson anemometer, whose axis of rotation is vertical, the vane anemometer must have its axis parallel to the direction of the wind and is therefore horizontal. Furthermore, since the wind varies in direction and the axis has to follow its changes, a wind vane or some other contrivance to fulfill the same purpose must be employed. A vane anemometer thus combines a propeller and a tail on the same axis to obtain accurate and precise wind speed and direction measurements from the same instrument. The speed of the fan is measured by a rev counter and converted to a windspeed by an electronic chip. Hence, volumetric flow rate may be calculated if the cross-sectional area is known. In cases where the direction of the air motion is always the same, as in ventilating shafts of mines and buildings, wind vanes known as air