the laser light, which is used to calculate the speed of the particles, and therefore the air around the anemometer. Ultrasonic anemometers Ultrasonic anemometers, first developed in the 1950s, use ultrasonic sound waves to measure wind velocity. They measure wind speed based on the time of flight of sonic pulses between pairs of transducers. Measurements from pairs of transducers can be combined to yield a measurement of velocity in 1-, 2-, or 3-dimensional flow. The spatial resolution is given by the path length between transducers, which is typically 10 to 20 cm. Ultrasonic anemometers can take measurements with very fine temporal resolution, 20 Hz or better, which makes them well suited for turbulence measurements. The lack of moving parts makes them appropriate for long-term use in exposed automated weather stations and weather buoys where the accuracy and reliability of traditional cup-and-vane anemometers are adversely affected by salty air or dust. Their main disadvantage is the distortion of the air flow by the structure supporting the transducers, which requires a correction based upon wind tunnel measurements to minimize the effect. An international standard for this process, ISO 16622 Meteorology—Ultrasonic anemometers/thermometers—Acceptance test methods for mean wind measurements is in general circulation. Another disadvantage is lower accuracy due to precipitation, where rain drops may vary the speed of sound. Since the speed of sound varies with temperature, and is virtually stable with pressure change, ultrasonic anemometers are also used as thermometers. Two-dimensional (wind speed and wind direction) sonic anemometers are used in applications such as weather stations, ship navigation, aviation, weather buoys and wind turbines. Monitoring wind turbines usually requires a refresh rate of wind speed measurements of 3 Hz, easily achieved by sonic anemometers. Three-dimensional sonic anemometers are widely used to measure gas emissions and ecosystem fluxes using the eddy covariance method when used with fast-response infrared gas analyzers or laser-based analyzers. Two-dimensional wind sensors are of two types: Two ultrasounds paths: These sensors have four arms. The disadvantage of this type of sensor is that when the wind comes in the direction of an ultrasound path, the arms disturb the airflow, reducing the accuracy of the resulting measurement. Three ultrasounds paths: These sensors have three arms. They give one path redundancy of the measurement which improves the sensor accuracy and reduces aerodynamic turbulence. Acoustic resonance anemometers Acoustic resonance anemometers are a more recent variant of sonic anemometer. The technology was invented