iodide reaction for glucose and tetrabromophenol blue reaction for the protein BSA). The bottom of the paper device is inserted into a sample solution prepared in-lab, and the amount of color change is observed. More recently, a paper-based microfluidic device using colorimetric detection was developed to quantify glucose in blood plasma. Blood plasma is separated from whole blood samples on a wax-printed device, where red blood cells are agglutinated by antibodies and the blood plasma is able to flow to a second compartment for the color-change reaction. Electrochemical detection has also been used in these devices. It provides greater sensitivity in quantification, whereas colorimetric detection is primarily used for qualitative assessments. Screen-printed electrodes and electrodes directly printed on filter paper have been used. One example of a paper-based microfluidic device utilizing electrochemical detection has a dumbbell shape to isolate plasma from whole blood. The current from the hydrogen peroxide produced in the aforementioned catalytic cycle is measured and converted into concentration of glucose. 3D devices for glucose detection Whitesides’ group also developed a 3D paper-based microfluidic device for glucose detection that can produce calibration curves on-chip because of the improved fluid flow design. This 3D device consists of layers of paper patterned with microfluidic channels that are connected by layers of double-sided adhesive tape with holes. The holes in the tape permit flow between channels in alternating layers of paper, so this device allows for more complicated flow paths and enables the detection of multiple samples in a large number (up to ~1,000) of detection zones in the last layer of paper. More recently, 3D paper-based microfluidic devices assembled using origami were developed. Unlike Whitesides’ design, these devices utilize a single layer of patterned paper that is then folded into multiple layers before sample solution is injected into the device. Subsequently, the device can be unfolded, and each layer of the device can be analyzed for the simultaneous detection of multiple analytes. This device is simpler and less expensive to fabricate than the aforementioned device using multiple layers of paper. Mixing between the channels in the different layers was not an issue in either device, so both devices were successful in quantifying glucose and BSA in multiple samples simultaneously. Environmental and food safety tests Paper-based microfluidic devices have several applications outside of the medical field. For example, paper-based biosensors have been used extensively in environmental monitoring. Two recent devices were