an interface for micro paper-based analytical devices μPAD and mass spectrometry. The technique, first described by Graham Cooks group at Purdue, involves applying a voltage to a triangular sheet of wet paper near the inlet of a mass spectrometer. Although the exact mechanism is not well understood, two modes of operation can occur: a multicone spray at high flow rates, and a single cone spray that occurs when solvent has been depleted. This is part of a larger effort to combine complex microfluidic manipulations with mass spectral detection. Wax printing hydrophobic barriers is a common method for creating distinct flow channels within paper devices, and this has been extended to μPAD-MS to enhance ionization efficiency (by enabling focusing of the analyte stream) and enable reaction mixing by wax printing on the triangular paper surface. Chromatographic separations have also been demonstrated on μPADs prior to paper-spray detection. Initially, paper-spray ionization was applied for the detection of small molecules, such as pharmaceuticals and drugs of abuse. However, it has also been shown that paper-spray ionization can ionize large proteins while retaining non-covalent interactions. Separation methods Few analytical detectors are truly specific for a single species; therefore some type of separation step is often necessary prior to detection. Moreover, separation allows for detection of multiple analytes within a single platform. Separations based upon planar chromatography (TLC) are perhaps the easiest to implement, since many μPADs are constructed with chromatographic paper. Typically, the separation channel is defined by wax-printing two hydrophobic barriers. Electrochemical detection is perhaps most common, likely due to its ease of implementation, although colorimetry, chemiluminscence, and mass spectral detection have also been used in conjunction with paper-based chromatographic separations. Despite the ease of implementation, planar chromatography is hindered by relatively low plate height (i.e., poor separation efficiency). Since the Chakraborty group demonstrated the feasibility of electrokinetic flow on μPADs, several applications of electrophoretic separations on μPADs have appeared in the literature. The Crooks group at UT-Austin successfully demonstrated that electrophoretic separations on μPADs could be accomplished at relatively low applied voltages compared to conventional electrophoretic devices due to the high field strengths that can be generated on very thin (180 μm) sheets of origami paper. Simpler separation methods can also be used on μPADs, for instance, the Henry group demonstrated the separation of plasma from whole blood using blood separation membranes. Flow control There are various ways to control the fluid