is used to measure samples. The cuvette is filled with sample, light is passed through the sample and intensity readings are taken. The slope spectroscopy technique can be applied using the same methods as in absorption spectroscopy. With the advent of accurate linear stages, variable pathlength absorption spectroscopy is easily applied experimentally. Other experimental methods include using ratios of slopes to build extinction coefficient spectra. This is possible because application of slope spectroscopy allows the scientist to keep concentration levels constant and vary path lengths. Background subtraction Variable pathlength absorption spectroscopy uses a determined slope to calculate concentration. As stated above this is a product of the molar absorptivity and the concentration. Since the actual absorbance value is taken at many data points at equal intervals, background subtraction is generally unnecessary. The image on the right is a linear plot showing both the background corrected data and the raw data. This shows that the absorbance values on the plot are offset by an equal amount and the slope of the two plots are equal. Thus, the concentration calculated from the two plots is equal. Other scalar components that contribute to the absorbance of a given sample like contaminants on the cuvette or a different cuvette material also are averaged out during the slope measurement. The technique is also applicable for in line measurements for TFF and chromatography applications. See also Applied spectroscopy References Further reading Scott Huffman, Keyur Soni and Joe Ferraiolo UV-Vis Based Determination of Protein Concentration: Validating and Implementing Slope Measurements Using Variable Pathlength Technology by September 2014 http://www.bioprocessintl.com/manufacturing/antibody-non-antibody/uv-vis-based-determination-protein-concentration-validating-implementing-slope-measurements-using-variable-pathlength-technology/ External links Absorption spectroscopy