Medtronic MiniMed 508 and Paradigm Series Insulin Pumps
The affected insulin pumps are designed to communicate using a wireless RF with other devices, such as blood glucose meters, glucose sensor transmitters, and CareLink USB devices. This wireless RF communication protocol does not properly implement authentication or authorization. An attacker with adjacent access to one of the affected insulin pump models can inject, replay, modify, and/or intercept data. This vulnerability could also allow attackers to change pump settings and control insulin delivery.CVE-2019-10964 has been assigned to this vulnerability. A CVSS v3 base score of 7.1 has been calculated; the CVSS vector string is (AV:A/AC:H/PR:N/UI:N/S:U/C:L/I:H/A:H).