Medtronic MyCareLink 24950 Patient Monitor
The affected product 's update service does not sufficiently verify the authenticity of the data uploaded. An attacker who obtains per-product credentials from the monitor and paired implantable cardiac device information can potentially upload invalid data to the Medtronic CareLink network.CVE-2018-10626 has been assigned to this vulnerability. A CVSS v3 base score of 4.4 has been calculated; the CVSS vector string is (AV:A/AC:H/PR:L/UI:N/S:C/C:L/I:L/A:N). The affected products use per-product credentials that are stored in a recoverable format. An attacker can use these credentials for network authentication and encryption of local data at rest.CVE-2018-10622 has been assigned to this vulnerability. A CVSS v3 base score of 4.9 has been calculated; the CVSS vector string is (AV:P/AC:H/PR:N/UI:N/S:C/C:H/I:N/A:N).