Medtronic MyCareLink Smart
The affected products contain an authentication protocol vulnerability where the method used to authenticate between the MCL Smart Patient Reader and the Medtronic MyCareLink Smart mobile app is vulnerable to bypass. This vulnerability enables an attacker to use another mobile device or malicious application on the patient 's smartphone to authenticate to the patient 's Medtronic Smart Reader, fooling the device into believing it is communicating with the original Medtronic smart phone application when executed within range of Bluetooth communication.CVE-2020-25183 has been assigned to this vulnerability. A CVSS v3 base score of 8.0 has been calculated; the CVSS vector string is (AV:A/AC:L/PR:N/UI:R/S:U/C:H/I:H/A:H). The affected products are vulnerable when an authenticated attacker runs a debug command, which can be sent to the patient reader and cause a heap overflow event within the MCL Smart Patient Reader software stack. The heap overflow could allow an attacker to remotely execute code on the MCL Smart Patient Reader, potentially leading to control of the deviceCVE-2020-25187 has been assigned to this vulnerability. A CVSS v3 base score of 8.8 has been calculated; the CVSS vector string is (AV:A/AC:L/PR:N/UI:R/S:C/C:H/I:H/A:H). The affected products are vulnerable to a race condition in the MCL Smart Patient Reader software update system, which allows unsigned firmware to be uploaded and executed on the Patient Reader. If exploited, an attacker could remotely execute code on the MCL Smart Patient Reader device, leading to control of the device.CVE-2020-27252 has been assigned to this vulnerability. A CVSS v3 base score of 8.8 has been calculated; the CVSS vector string is (AV:A/AC:L/PR:N/UI:R/S:C/C:H/I:H/A:H).