Philips Patient Monitoring Devices (Update C)
In Patient Information Center iX (PICiX) Versions B.02, C.02, C.03, the software saves user-provided information into a comma-separated value (CSV) file, but it does not neutralize or incorrectly neutralizes special elements that could be interpreted as a command when the file is opened by spreadsheet software. In Patient Information Center iX (PICiX) Versions B.02, C.02, C.03, the software does not neutralize or incorrectly neutralizes user-controllable input before it is placed in output that is then used as a webpage and served to other users. Successful exploitation could lead to unauthorized access to patient data via a read-only web application. In Patient Information Center iX (PICiX) Version B.02, C.02, C.03, and PerformanceBridge Focal Point Version A.01, when an actor claims to have a given identity, the software does not prove or insufficiently proves the claim is correct. In Patient Information Center iX (PICiX) Versions C.02 and C.03, PerformanceBridge Focal Point Version A.01, IntelliVue patient monitors MX100, MX400-MX550, MX750, MX850, and IntelliVue X3 Versions N and prior, the software does not check or incorrectly checks the revocation status of a certificate, which may cause it to use a compromised certificate. In Patient Information Center iX (PICiX) Versions C.02, C.03, the software parses a formatted message or structure but does not handle or incorrectly handles a length field that is inconsistent with the actual length of the associated data, causing the application on the surveillance station to restart. In Patient Information Center iX (PICiX) Versions C.02, C.03, PerformanceBridge Focal Point Version A.01, the product receives input that is expected to be well-formed (i.e., to comply with a certain syntax) but it does not validate or incorrectly validates that the input complies with the syntax, causing the certificate enrollment service to crash. It does not impact monitoring but prevents new devices from enrolling. In IntelliVue patient monitors MX100, MX400-550, MX600, MX700, MX750, MX800, MX850, MP2-MP90, and IntelliVue X2 and X3 Versions N and prior, the product receives input or data but does not validate or incorrectly validates that the input has the properties required to process the data safely and correctly, which can induce a denial-of-service condition through a system restart. In Patient Information Center iX (PICiX) Versions B.02, C.02, C.03, the product exposes a resource to the wrong control sphere, providing unintended actors with inappropriate access to the resource. The application on the surveillance station operates in kiosk mode, which is vulnerable to local breakouts that could allow an attacker with physical access to escape the restricted environment with limited privileges.