Philips Patient Information Center iX (PIC iX) and Efficia CM Series
Patient Information Center iX (PIC iX) Versions C.02 and C.03 receives input or data, but does not validate or incorrectly validates that the input has the properties required to process the data safely and correctly. CVE-2021-43548 has been assigned to this vulnerability. A CVSS v3 base score of 6.5 has been calculated; the CVSS vector string is (AV:A/AC:L/PR:N/UI:N/S:U/C:N/I:N/A:H). The use of a hard-coded cryptographic key significantly increases the possibility encrypted data may be recovered from the Patient Information Center iX (PIC iX) Versions B.02, C.02, and C.03. CVE-2021-43552 has been assigned to this vulnerability. A CVSS v3 base score of 6.1 has been calculated; the CVSS vector string is (AV:L/AC:L/PR:L/UI:N/S:U/C:H/I:L/A:N). The use of a broken or risky cryptographic algorithm is an unnecessary risk that may result in the exposure of sensitive information, which affects the communications between Patient Information Center iX (PIC iX) Versions C.02 and C.03 and Efficia CM Series Revisions A.01 to C.0x and 4.0. CVE-2021-43550 has been assigned to this vulnerability. A CVSS v3 base score of 5.9 has been calculated; the CVSS vector string is (AV:A/AC:H/PR:N/UI:N/S:U/C:H/I:L/A:N).