Smiths Medical Medfusion 4000 Wireless Syringe Infusion Pump Vulnerabilities (Update A)
A third-party component used in the pump does not verify input buffer size prior to copying, leading to a buffer overflow, allowing remote code execution on the target device. The pump receives the potentially malicious input infrequently and under certain conditions, increasing the difficulty of exploitation. CVE-2017-12718 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). A third-party component used in the pump does not verify input buffer size prior to copying, leading to a buffer overflow, allowing remote code execution on the target device. The pump receives the potentially malicious input infrequently and under certain conditions, increasing the difficulty of exploitation. CVE-2017-12718 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). A third-party component used in the pump reads memory out of bounds, causing the communications module to crash. Smiths Medical assesses that the crash of the communications module would not impact the operation of the therapeutic module. CVE-2017-12722 been assigned to this vulnerability. A CVSS v3 base score of 5.3 has been assigned; the CVSS vector string is (AV:N/AC:L/PR:N/UI:N/S:U/C:N/I:N/A:L). A third-party component used in the pump does not verify input buffer size prior to copying, leading to a buffer overflow, allowing remote code execution on the target device. The pump receives the potentially malicious input infrequently and under certain conditions, increasing the difficulty of exploitation. CVE-2017-12718 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). A third-party component used in the pump reads memory out of bounds, causing the communications module to crash. Smiths Medical assesses that the crash of the communications module would not impact the operation of the therapeutic module. CVE-2017-12722 been assigned to this vulnerability. A CVSS v3 base score of 5.3 has been assigned; the CVSS vector string is (AV:N/AC:L/PR:N/UI:N/S:U/C:N/I:N/A:L). The pump with default network configuration uses hard-coded credentials to automatically establish a wireless network connection. The pump will establish a wireless network connection even if the pump is Ethernet connected and active; however, if the wireless association is established and the Ethernet cable is attached, the pump does not attach the network stack to the wireless network. In this scenario, all network traffic is instead directed over the wired Ethernet connection. CVE-2017-12725 been assigned to this vulnerability. A CVSS v3 base score of 9.8 has been assigned; the CVSS vector string is (AV:N/AC:L/PR:N/UI:N/S:U/C:H/I:H/A:H). A third-party component used in the pump does not verify input buffer size prior to copying, leading to a buffer overflow, allowing remote code execution on the target device. The pump receives the potentially malicious input infrequently and under certain conditions, increasing the difficulty of exploitation. CVE-2017-12718 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). A third-party component used in the pump reads memory out of bounds, causing the communications module to crash. Smiths Medical assesses that the crash of the communications module would not impact the operation of the therapeutic module. CVE-2017-12722 been assigned to this vulnerability. A CVSS v3 base score of 5.3 has been assigned; the CVSS vector string is (AV:N/AC:L/PR:N/UI:N/S:U/C:N/I:N/A:L). The pump with default network configuration uses hard-coded credentials to automatically establish a wireless network connection. The pump will establish a wireless network connection even if the pump is Ethernet connected and active; however, if the wireless association is established and the Ethernet cable is attached, the pump does not attach the network stack to the wireless network. In this scenario, all network traffic is instead directed over the wired Ethernet connection. CVE-2017-12725 been assigned to this vulnerability. A CVSS v3 base score of 9.8 has been assigned; the CVSS vector string is (AV:N/AC:L/PR:N/UI:N/S:U/C:H/I:H/A:H). The FTP server on the pump does not require authentication if the pump is configured to allow FTP connections. CVE-2017-12720 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). A third-party component used in the pump does not verify input buffer size prior to copying, leading to a buffer overflow, allowing remote code execution on the target device. The pump receives the potentially malicious input infrequently and under certain conditions, increasing the difficulty of exploitation. CVE-2017-12718 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). A third-party component used in the pump reads memory out of bounds, causing the communications module to crash. Smiths Medical assesses that the crash of the communications module would not impact the operation of the therapeutic module. CVE-2017-12722 been assigned to this vulnerability. A CVSS v3 base score of 5.3 has been assigned; the CVSS vector string is (AV:N/AC:L/PR:N/UI:N/S:U/C:N/I:N/A:L). The pump with default network configuration uses hard-coded credentials to automatically establish a wireless network connection. The pump will establish a wireless network connection even if the pump is Ethernet connected and active; however, if the wireless association is established and the Ethernet cable is attached, the pump does not attach the network stack to the wireless network. In this scenario, all network traffic is instead directed over the wired Ethernet connection. CVE-2017-12725 been assigned to this vulnerability. A CVSS v3 base score of 9.8 has been assigned; the CVSS vector string is (AV:N/AC:L/PR:N/UI:N/S:U/C:H/I:H/A:H). The FTP server on the pump does not require authentication if the pump is configured to allow FTP connections. CVE-2017-12720 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). The FTP server on the pump contains hardcoded credentials, which are not fully initialized. The FTP server is only accessible if the pump is configured to allow FTP connections. CVE-2017-12724 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). A third-party component used in the pump does not verify input buffer size prior to copying, leading to a buffer overflow, allowing remote code execution on the target device. The pump receives the potentially malicious input infrequently and under certain conditions, increasing the difficulty of exploitation. CVE-2017-12718 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). A third-party component used in the pump reads memory out of bounds, causing the communications module to crash. Smiths Medical assesses that the crash of the communications module would not impact the operation of the therapeutic module. CVE-2017-12722 been assigned to this vulnerability. A CVSS v3 base score of 5.3 has been assigned; the CVSS vector string is (AV:N/AC:L/PR:N/UI:N/S:U/C:N/I:N/A:L). The pump with default network configuration uses hard-coded credentials to automatically establish a wireless network connection. The pump will establish a wireless network connection even if the pump is Ethernet connected and active; however, if the wireless association is established and the Ethernet cable is attached, the pump does not attach the network stack to the wireless network. In this scenario, all network traffic is instead directed over the wired Ethernet connection. CVE-2017-12725 been assigned to this vulnerability. A CVSS v3 base score of 9.8 has been assigned; the CVSS vector string is (AV:N/AC:L/PR:N/UI:N/S:U/C:H/I:H/A:H). The FTP server on the pump does not require authentication if the pump is configured to allow FTP connections. CVE-2017-12720 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). The FTP server on the pump contains hardcoded credentials, which are not fully initialized. The FTP server is only accessible if the pump is configured to allow FTP connections. CVE-2017-12724 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). Telnet on the pump uses hardcoded credentials, which can be used if the pump is configured to allow external communications. Smiths Medical assesses that it is not possible to upload files via Telnet and the impact of this vulnerability is limited to the communications module. CVE-2017-12726 been assigned to this vulnerability. A CVSS v3 base score of 5.6 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:L/I:L/A:L). A third-party component used in the pump does not verify input buffer size prior to copying, leading to a buffer overflow, allowing remote code execution on the target device. The pump receives the potentially malicious input infrequently and under certain conditions, increasing the difficulty of exploitation. CVE-2017-12718 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). A third-party component used in the pump reads memory out of bounds, causing the communications module to crash. Smiths Medical assesses that the crash of the communications module would not impact the operation of the therapeutic module. CVE-2017-12722 been assigned to this vulnerability. A CVSS v3 base score of 5.3 has been assigned; the CVSS vector string is (AV:N/AC:L/PR:N/UI:N/S:U/C:N/I:N/A:L). The pump with default network configuration uses hard-coded credentials to automatically establish a wireless network connection. The pump will establish a wireless network connection even if the pump is Ethernet connected and active; however, if the wireless association is established and the Ethernet cable is attached, the pump does not attach the network stack to the wireless network. In this scenario, all network traffic is instead directed over the wired Ethernet connection. CVE-2017-12725 been assigned to this vulnerability. A CVSS v3 base score of 9.8 has been assigned; the CVSS vector string is (AV:N/AC:L/PR:N/UI:N/S:U/C:H/I:H/A:H). The FTP server on the pump does not require authentication if the pump is configured to allow FTP connections. CVE-2017-12720 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). The FTP server on the pump contains hardcoded credentials, which are not fully initialized. The FTP server is only accessible if the pump is configured to allow FTP connections. CVE-2017-12724 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). Telnet on the pump uses hardcoded credentials, which can be used if the pump is configured to allow external communications. Smiths Medical assesses that it is not possible to upload files via Telnet and the impact of this vulnerability is limited to the communications module. CVE-2017-12726 been assigned to this vulnerability. A CVSS v3 base score of 5.6 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:L/I:L/A:L). The pump does not validate host certificate, leaving the pump vulnerable to a man-in-the-middle (MITM) attack. CVE-2017-12721 been assigned to this vulnerability. A CVSS v3 base score of 7.5 has been assigned; the CVSS vector string is (AV:N/AC:L/PR:N/UI:N/S:U/C:H/I:N/A:N). A third-party component used in the pump does not verify input buffer size prior to copying, leading to a buffer overflow, allowing remote code execution on the target device. The pump receives the potentially malicious input infrequently and under certain conditions, increasing the difficulty of exploitation. CVE-2017-12718 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). A third-party component used in the pump reads memory out of bounds, causing the communications module to crash. Smiths Medical assesses that the crash of the communications module would not impact the operation of the therapeutic module. CVE-2017-12722 been assigned to this vulnerability. A CVSS v3 base score of 5.3 has been assigned; the CVSS vector string is (AV:N/AC:L/PR:N/UI:N/S:U/C:N/I:N/A:L). The pump with default network configuration uses hard-coded credentials to automatically establish a wireless network connection. The pump will establish a wireless network connection even if the pump is Ethernet connected and active; however, if the wireless association is established and the Ethernet cable is attached, the pump does not attach the network stack to the wireless network. In this scenario, all network traffic is instead directed over the wired Ethernet connection. CVE-2017-12725 been assigned to this vulnerability. A CVSS v3 base score of 9.8 has been assigned; the CVSS vector string is (AV:N/AC:L/PR:N/UI:N/S:U/C:H/I:H/A:H). The FTP server on the pump does not require authentication if the pump is configured to allow FTP connections. CVE-2017-12720 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). The FTP server on the pump contains hardcoded credentials, which are not fully initialized. The FTP server is only accessible if the pump is configured to allow FTP connections. CVE-2017-12724 been assigned to this vulnerability. A CVSS v3 base score of 8.1 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:H/I:H/A:H). Telnet on the pump uses hardcoded credentials, which can be used if the pump is configured to allow external communications. Smiths Medical assesses that it is not possible to upload files via Telnet and the impact of this vulnerability is limited to the communications module. CVE-2017-12726 been assigned to this vulnerability. A CVSS v3 base score of 5.6 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:L/I:L/A:L). The pump does not validate host certificate, leaving the pump vulnerable to a man-in-the-middle (MITM) attack. CVE-2017-12721 been assigned to this vulnerability. A CVSS v3 base score of 7.5 has been assigned; the CVSS vector string is (AV:N/AC:L/PR:N/UI:N/S:U/C:H/I:N/A:N). The pump stores some passwords in the configuration file, which are accessible if the pump is configured to allow external communications. CVE-2017-12723 been assigned to this vulnerability. A CVSS v3 base score of 3.7 has been assigned; the CVSS vector string is (AV:N/AC:H/PR:N/UI:N/S:U/C:L/I:N/A:N).