text: The technology works by probing an optical standing wave, or the sum of the standing waves in the case of polychromatic light, created by a light to be analyzed. In a SWIFTS linear configuration (true Lippman configuration), the stationary wave is created by a single-mode waveguide ended by a fixed mirror. The stationary wave is regularly sampled on one side of a waveguide using nano-scattering dots. These dots are located in the evanescent field. These nanodots are characterized by an optical index difference with the medium in which the evanescent field is located. The light is then scattered around an axis perpendicular to the waveguide. For each dot, this scattered light is detected by a pixel aligned with this axis. The intensity detected is therefore proportional to the intensity inside the waveguide at the exact location of the dot. This results in a linear image of the interferogram. No moving parts are used. A mathematical function known as a Lippmann transform, similar to a Fourier transform, is then applied to this linear image and gives the spectrum of the light.
  The interferogram is truncated. Only the frequencies corresponding to the zero optical path difference at the mirror, up to the farthest dots are sampled. Higher frequencies are rejected. This interferogram’s truncation determines the spectral resolution. The interferogram is undersampled. A consequence of this under-sampling is a limitation of the wavelength bandwidth to which the mathematical function is applied.
  SWIFTS technology displays the Fellgetts advantage, which is derived from the fact that an interferometer measures wavelengths simultaneously with the same elements of the detector, whereas a dispersive spectrometer measures them successively. Fellgetts advantage also states that when collecting a spectrum whose measurement noise is dominated by detector noise, a multiplex spectrometer such as a Fourier-transform spectrometer will produce a relative improvement in the signal-to-noise ratio, with respect to an equivalent scanning monochromator, that is approximately equal to the square root of the number of sample points comprising the spectrum. The Connes advantage states that the wavenumber scale of an interferometer, derived from a helium–neon laser, is more accurate and boasts better long-term stability than the calibration of dispersive instruments.