text: The AFM-IR technique can simultaneously provide complementary measurements of the mechanical stiffness and dissipation of a sample surface. When infrared light is absorbed by the sample the resulting rapid thermal expansion excites a "contact resonance" of the AFM cantilever, i.e. a coupled resonance resulting from the properties of both the cantilever and the stiffness and damping of the sample surface. Specifically, the resonance frequency shifts to higher frequencies for stiffer materials and to lower frequencies for softer material. Additionally, the resonance becomes broader for materials with larger dissipation. These contact resonances have been studied extensively by the AFM community (see, for example, atomic force acoustic microscopy). Traditional contact resonance AFM requires an external actuator to excite the cantilever contact resonances. In AFM-IR these contact resonances are automatically excited every time an infrared pulse is absorbed by the sample. So the AFM-IR technique can measure the infrared absorption by the amplitude of the cantilever oscillation response and the mechanical properties of the sample via the contact resonance frequency and quality factor.