text: The NICE-OHMS technique combines cavity enhanced absorption spectrometry (CEAS) for prolonged interaction length with the sample with frequency modulation (fm) spectrometry FMS for reduction of 1/f noise. By choosing the fm-modulation frequency equal to the free spectral range (FSR) of the cavity, all components of the spectral fm-triplet are transmitted through the cavity in an identical manner. Therefore, the cavity does not compromise the balance of the fm-triplet, which otherwise would give rise to fm-background signals. It also does not convert any fluctuations of the laser frequency with respect to the transmission mode of the cavity to intensity modulation, which would deteriorate the detectability by the introduction of intensity noise. This is referred to as "noise immunity". All this implies that FMS can be performed as if the cavity were not present, yet fully benefiting from the prolonged interaction length.