text: This technique (named after their inventors) is based on the line-broadening and self-reversal of emission lines from HCL when high current is applied. Total absorption is measured with normal lamp current, i.e., with a narrow emission line, and background absorption after application of a high-current pulse with the profile of the self-reversed line, which has little emission at the original wavelength, but strong emission on both sides of the analytical line. The advantage of this technique is that only one radiation source is used; among the disadvantages are that the high-current pulses reduce lamp lifetime, and that the technique can only be used for relatively volatile elements, as only those exhibit sufficient self-reversal to avoid dramatic loss of sensitivity. Another problem is that background is not measured at the same wavelength as total absorption, making the technique unsuitable for correcting structured background.