For the analyzed LIG samples, one must consider a different source of the ohmic drop between the WE and the CE electrodes, forcing an alternative approach, which may be performed based on manual correction after the measurements. Fig. compares the raw exemplar result with its compensated versions for the P12% sample. One methodology that can be applied for manual compensation of the ohmic drop is to assume that the potential "felt" by the working electrode V(t) is equal to its true potential E(t) minus the IR voltage drop, according to the equation ( ) :