 High vacuum chest drainage was once considered inadvisable because of a possible suction injury to soft tissues, such as the lungs or mediastinum. Wakabayashi showed that excessive suction force generated by a high vacuum pressure was related to the surface area of the suction applied to the tissues and not the vacuum pressure itself12. He postulated that suction injury to the lung would not occur if the suction force of a chest tube is below the capillary blood pressure of 30 - 35 torr2. Based on this theory, Wakabayashi developed a double-lumen chest drainage tube3 Trade name NO-NUMOTM and its clinical use was approved by FDA4 in 2002. A prototype 7-5Fr double-lumen chest tube was successfully tested in laboratory animals and three patients12 and the safety of high vacuum chest drainage was proven in a large number of patients undergoing uncomplicated thoracoscopic surgeries2. The unique design of the new drainage tube is the small diameter double lumen tube with hundreds of tiny holes that allows rapid drainage without damaging effects of high suction over a limited surface area. High vacuum suction is available on wall units already in place in the OR and ICU beds. To adopt this system for clinical use, a high vacuum gauge was needed most dont go beyond 200 torr to monitor suction, and a high pressure battery powered vacuum pump for transport was necessary, so that drainage was maintained and blood clotting in the tube avoided during the period of prolonged transport from the OR and the setup in the ICU. Several brands of portable vacuum pumps on the market were evaluated before concluding that the VarioTM pump Medela Inc. was superior. The maximal vacuum pressure that Vario pump can generate is 630 torr or 84 kPa which is equivalent to -857 cmH2O. 