 One of the contributing factors to the variability in outcomes amongst Cochlear Implant CI recipients was reported to be the placement of the electrode array in the scala tympani. It seems that the correct placement of the electrode initially into the scala tympani and subsequent avoidance of dislocation into the scala vestibuli as the insertion progresses, is a key factor in achieving good speech perception outcomes. Another important aspect related to the performance is the achievement of consistent electrical coverage with the electrode. Data reported for electrodes of different manufacturers give depths ranging from 240 - 600 degrees showing the considerable variation across subjects. The HiFocus mid scala electrode was developed to cover one and a quarter turn and with the pre-curved design to be less susceptible to variations in individual cochlea dimensions and insertion techniques. A further mechanical feature of the pre-curved design is the avoidance of forces against the cochlear lateral wall and associated lower susceptibility of the electrode for moving out of the cochlea following insertion. Recently, a cone beam CT CBCT technique is being explored in the field of ENT with the potential to overcome some of the issues associated with the conventional CT techniques such as scattering, radiation and low isometric resolution. Images with comparable details to those of e.g. micro CTs are possible with much lower radiation dose. Modern imaging software i.e. 3D Slicer may be used for 3D reconstruction, post processing and Brainsfit for registration. Registration is the alignment of two scans in the same coordinate system. This enables the identification of differences between two images recorded at time x and y. Once accurately superimposed any difference between the two images may be identified with an accuracy of 0.2 mm. Using these methods, one can assess the stability and the position of the electrode in the cochlea. 