 Instruments: The Ocular Response Analyzer ORA Reichert, INC, Depew, NY was used to assess the biomechanical properties of the cornea. This instrument utilizes a dynamic bi-directional applanation process to measure the biomechanical properties of cornea and estimate IOP. A rapid air pulse applies force to the cornea and an advanced electro-optical system monitors corneal deformation. A precisely-metered collimated-air-pulse pushes the cornea inwards, past a flat state and into a slight concavity. Milliseconds after applanation, the air pump shuts off and the pressure gradually declines. As the pressure decreases, the cornea first passes through an applanated state before resuming its original curved structure. The applanation detection system monitors the corneal movement throughout the entire process. Two separate pressure values are derived from the INWARD and OUTWARD applanation events. The difference between these two pressure values is termed CH. This biomechanical property reflects corneal absorption and dissipation of the energy from an applied force. Derived from the measurement of CH is the Corneal resistance factor CRF. CRF represents the corneas ability to resist deformation in the presence of an external force. 